Last post by admin - October 30, 2023, 09:50:26 AM
The ProTrack (USB only) and Scanner are still supported in the current version of Huntron Workstation 4.3. You can contact your local Huntron distributor and inquire about a license. https://huntron.com/sales-support/sales.htm
Last post by jvthorsley - October 30, 2023, 09:46:26 AM
I'll jump in before Curtis on this one. It all depends on what version of Windows you want to run Workstation on and do you want Tech support on it. I got the most recent version because as written about on this Forum I wanted to connect a Keysight DMM to the output of my Scanner. Plus a Keysight E4980AL LCR meter connected in the same way.
Last post by admin - October 25, 2023, 12:20:19 PM
This is mostly an update for the new Trackers (2800, 2800S and 3200S with serial number 3000 and above) that have the new version main board. We recommend users with these Tracker install this version which also includes a firmware update (use the Diagnostics under the Tools menu/Maintenance). We did address a relay selection issue for the ProTrack Scanner with this release as well. Here is a download link: Huntron Workstation 4.3.8705
Last post by admin - October 16, 2023, 11:22:39 AM
My thought is that power supply is starting to fail because when the leads are shorted in the LOW range there is a large current demand. Set the ProTrack to a range such as 3V, 1Kohms ( a much lower current demand) and short the leads. Replace the electrolytic caps as JVTHorsley suggested.
It appears that ProTrack has had some modifications. The white text, blue background LCD is definitely not original. It is also one of the first production run ProTrack Model 10 units based on the serial number. It was likely upgraded to a Model 20 later on. It also needs a very good cleaning!